IEC 62052-11, #7.2
Short-time overcurrent test shall not damage the meter. The meter shall perform correctly when back to its initial working condition.
Conditions for direct connected meters:
Test current: 30 Imax, duration one half cycle at nominal frequency
Conditions for meters connected trough current transformers:
Test current: 20 Imax, duration 0.5 s